1, Around and the bottom of the laboratory oven must be keep clean all the time. 2, Before the operation, impurity inside the machine should be sweep away; test space should be cleaned at least once a year 3, When the test time is over, the machine must be assured of turning off before take out the products. 4, Cooling system is the core of high and low temperature chamber, so we should check the cooper pipe of the leakage etc. Situation once half a year, other function like splice, weld, we must solve it immediately once the oil stain leaked. 5, Condenser should have a periodic maintenance, keep clean. If there’s dust on the condenser, the high- voltage switch of the compressor would trip and appear false alarm. So the condenser should be maintained once a month, sweep away the dust adhere to the condenser by using vacuum cleaner, or remove it by hard fur brush. 6, Every time when the experiment is finished, we should clean up the test chamber by clean water; And after cleaned up the lab oven, it should be drying, keep the environmental test chamber dry. 7, circuit breaker, overheat protector, testing items and safe protection of conductor should be checked regularly; Also, please read the instruction book carefully, so you can operate the machine correctly.
All posts by: admin
What’s the difference between temperature fluctuation and deviation of Temperature and humidity chamber？
Temperature and humidity chamber can imitate different temperature and humid environment, apply to test electrical, electronic, foodstuff, automobile, rubber, plastics, metal etc. products, this Temperature and humidity chamber meet GB/T2423, GJB150A1/4 reliability and stability etc. index in various kinds of severe environment, Temperature and humidity chamber can provide evidence of quality and reliability in predicting and improving products’ quality. Now what’s the difference between temperature fluctuation and deviation of this environmental test chamber? As follows, Temperature deviation is an allowance(qualified) range of reference value, for example, temperature deviation is 100（±2）℃, then when control the temperature, the temperature of the work space at 98-102 ℃ is qualified. While temperature fluctuation means, when set the standard value, the temperature in the chamber will appear certain fluctuation in short time because of the non-uniformity of emanating heat. The fluctuation range may be larger, such as ±5 ℃; that is to say, temperature in test chamber is fluctuate within 95–105 ℃.
3）temperature uniformity When the temperature is stable, the maximum of temperature average value’ different of two arbitrarily point in work space. The index is more suitable to check the core technology of products than the temperature departure index, so many enterprises are deliberately conceal this item for sample and scheme. Usually, the standard index is ≤2℃. But objectively speaking, clean high temperature test chamber is request to keep the uniformity ≤2℃ when the temperature reach to above 200℃, in fact that is uneconomic to customers 4) temperature deviation After the temperature is stable, the difference between the average value of center point in work space and the average value of other point in any time interval. Although old and new standard have the same identification and appellation to this index, the examination has kinda changed, the new standard is more real and strict, while the check time is shorter. Usually, the standard index is ±2℃, high temperature chamber can use ±2% when the temperature reach to above 200℃. In the taiwan index, there are no temperature uniformity and deviation as mainland of China called, they named it as distribution uniformity of temperature, sample index are ±0.5℃、±1℃、±1.5℃ etc. 5) temperature change rate This is an index to check the volume of product’s configuration. Generally, for customers that have no demand on temperature change, it’s ok to mainly understand the time from room temperature to extreme low-temperature. For those tests that request temperature change, the simplest way for customers is to sign tech agreement when ordering environmental test chambers. And when sign the agreement, you must know the loading test at first, then, list the volume, size, weight, basic material constitute and temperature deviation etc. index, and other test standards related to the temp spots, procedure description and temperature changing graphs from beginning to the end. By doing this, you can trustingly order laboratory ovens products.
1, Preface Although Environmental testing industry is a traditional industry that started not too late in China, getting truly universal is around the latest 20 years. Nowadays, there are more than 300 factories and companies in various scale in China, but very few enterprise is certainly understand and grasp the core and essence of products. For customers, products are just like seller’s market, they have many choices, but because few high-skilled users know the usage function of product, and in addition, the so-called pre-pay policy in transportation and sales operation in industry, and the distributors play game technical index, and so on customer’s tragic of history. But today we are not to talk about that bad history, I’m in technical work, and i’m teaching you guys to avoid being confused and conceived. So i made some explanations on frequently-used temperature and humidity index and common configuration, hope it will help users. Of course, there are many environmental test indication project, and configuration operation patterns are all has its advantages. 2 performance index explanation 1 temperature index 1)temperature range: temperature range refers to the ultimate temperature that lab oven can suffer or reach. Usually, constant conception that can be control should be a long-time stable operational […]
Company PSV Technologies,LLC offers engineering services and provides the deliveries of imported and domestic high-tech equipment for the industry-leading research centers, enterprises and educational institutions.